
New TopShim run:

version = topshim 4.4.0
date = 2026-04-21
time = 12:39:59.102
command = topshim tunebxyz tuneaz ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 6.3 Hz, Target= 0.2 Hz, After= 0.4 Hz
Iteration 2: FWHM Before= 0.4 Hz, Target= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.503


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	-129
Z2	-28
Z3	-85
Z4	-130
Z5	-61

duration = 1 min 4 sec
completed successfully
finished Tue Apr 21 12:41:02 2026


========================================================================
