
New TopShim run:

version = topshim 4.4.0
date = 2025-09-01
time = 14:14:54.783
command = topshim 1d ss coil=1H ss tuneaoff selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = Acetone
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 2.04 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.2 Hz, Target= 0.3 Hz, After= 0.7 Hz
Iteration 2: FWHM Before= 0.7 Hz, Target= 0.7 Hz, After= 0.7 Hz
Iteration 3: FWHM Before= 0.7 Hz, Target= 0.4 Hz, After= 0.4 Hz
Iteration 4: FWHM Before= 0.4 Hz, Target= 0.5 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.586

shim changes: 
Z	+83
Z2	+4
Z3	-121
Z4	+71
Z5	-43
Z6	-255

duration = 1 min 15 sec
completed successfully
finished Mon Sep  1 14:16:09 2025


========================================================================

New TopShim run:

version = topshim 4.4.0
date = 2025-09-01
time = 14:29:40.121
command = topshim 1d ss coil=1H ss tuneaoff selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = D2O
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 4.70 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 3.6 Hz, Target= 0.5 Hz, After= 0.6 Hz
Iteration 2: FWHM Before= 0.6 Hz, Target= 0.3 Hz, After= 0.4 Hz
Iteration 3: FWHM Before= 0.4 Hz, Target= 0.2 Hz, After= 0.4 Hz
Iteration 4: FWHM Before= 0.4 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.424

shim changes: 
Z	-173
Z2	+8
Z3	+4
Z4	-398
Z5	-86
Z6	+955

duration = 1 min 8 sec
completed successfully
finished Mon Sep  1 14:30:47 2025


========================================================================
