
New TopShim run:

version = topshim 4.4.0
date = 2025-09-01
time = 14:20:13.116
command = topshim 1d ss coil=1H ss tuneaoff selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = D2O
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 4.70 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 1.2 Hz, Target= 0.4 Hz, After= 0.4 Hz
Iteration 2: FWHM Before= 0.4 Hz, Target= 0.4 Hz, After= 0.5 Hz
Iteration 3: FWHM Before= 0.5 Hz, Target= 0.4 Hz, After= 0.4 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.515

shim changes: 
Z	+28
Z2	+11
Z3	-54
Z4	-87
Z5	+122
Z6	+228

duration = 59 sec
completed successfully
finished Mon Sep  1 14:21:11 2025


========================================================================
