
New TopShim run:

version = topshim 4.4.0
date = 2025-08-22
time = 10:07:31.418
command = topshim 1d ls coil=1H ls tuneaoff lockoff 1h o1p=4.26 selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.26 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 2.2 Hz, Target= 0.2 Hz, After= 0.6 Hz
Iteration 2: FWHM Before= 0.6 Hz, Target= 0.2 Hz, After= 0.4 Hz
Iteration 3: FWHM Before= 0.4 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.342

shim changes: 
Z	+59
Z2	-12
Z3	+16
Z4	+66
Z5	+104
Z6	-346

duration = 19 sec
completed successfully
finished Fri Aug 22 10:07:49 2025


========================================================================
