
New TopShim run:

version = topshim 4.4.0
date = 2025-08-25
time = 15:09:47.389
command = topshim 1d ls coil=1H ls tuneaoff lockoff 1h o1p=4.78 selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = CDCl3
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 4.78 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 4.2 Hz, Target= 0.2 Hz, After= 0.7 Hz
Iteration 2: FWHM Before= 0.7 Hz, Target= 0.2 Hz, After= 0.3 Hz
Iteration 3: FWHM Before= 0.3 Hz, Target= 0.2 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.366

shim changes: 
Z	-200
Z2	-29
Z3	-73
Z4	-167
Z5	-143
Z6	+487

duration = 18 sec
completed successfully
finished Mon Aug 25 15:10:05 2025


========================================================================
