
New TopShim run:

version = topshim 4.4.0
date = 2026-02-04
time = 12:09:47.683
command = topshim 1d ls coil=1H ls tuneaoff lockoff 1h o1p=7.06 selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = D2O
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 7.06 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.2 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration 2: FWHM Before= 0.6 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.364

shim changes: 
Z	-35
Z2	+26
Z3	-39
Z4	-7
Z5	+19
Z6	+375

duration = 14 sec
completed successfully
finished Wed Feb  4 12:10:01 2026


========================================================================
