
New TopShim run:

version = topshim 4.4.0
date = 2026-02-03
time = 14:37:01.773
command = topshim 1d ls coil=1H ls tuneaoff lockoff 1h o1p=7.2 selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = CDCl3
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 7.20 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.5 Hz, Target= 0.2 Hz, After= 0.2 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.316

shim changes: 
Z	-12
Z2	+13
Z3	-21
Z4	+81
Z5	-69
Z6	+136

duration = 10 sec
completed successfully
finished Tue Feb  3 14:37:11 2026


========================================================================
