
New TopShim run:

version = topshim 4.4.0
date = 2026-02-11
time = 13:19:17.316
command = topshim 1d ls coil=1H ls tuneaoff lockoff 1h o1p=6.6 selwid=0.5 ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = DMSO
shim nucleus = 1H
nucleus optimised for = 1H
o1p (set) = 6.60 ppm
selective excitation width = 0.5 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 0.7 Hz, Target= 0.3 Hz, After= 0.3 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.409

shim changes: 
Z	-13
Z2	+4
Z3	-1
Z4	+81
Z5	-3
Z6	-189

duration = 10 sec
completed successfully
finished Wed Feb 11 13:19:26 2026


========================================================================
