
New TopShim run:

version = topshim 4.4.0
date = 2025-10-25
time = 15:30:51.696
command = topshim tunebxyz tuneaz ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before=24.2 Hz, Target= 0.3 Hz, After= 0.6 Hz
Iteration 2: FWHM Before= 0.6 Hz, Target= 0.7 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.731


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	-339
Z2	+50
Z3	-11
Z4	+28
Z5	-28

duration = 1 min 9 sec
completed successfully
finished Sat Oct 25 15:32:00 2025


========================================================================
