
New TopShim run:

version = topshim 4.4.0
date = 2026-04-08
time = 13:41:09.610
command = topshim tunebxyz tuneaz ordmax=6

FINE SHIMMING

Parameters:

probe = Z113652_0123
solvent = CDCl3
shim nucleus = 2H
nucleus optimised for = 1H
o1p (from lock) = 7.24 ppm

Results:


1D Shimming:
Iteration 1: FWHM Before= 1.0 Hz, Target= 0.5 Hz, After= 0.5 Hz
Shim succeeded

Lineshape quality factor (1D spectrum, spinning) = 0.639


Performing tune after shimming
tune lock level increase = 1.0

shim changes: 
Z	+84
Z2	+2
Z3	-28
Z4	+177
Z5	+53
Z6	-540

duration = 1 min 4 sec
completed successfully
finished Wed Apr  8 13:42:13 2026


========================================================================
